Here is nébuliseur of table, Sprai for Airborne disinfection small size autonome, effective for small volumes such as psm Microbiological Safety Station.
Orientable et programmable code> one, exists in 400, 800,1200 ml par heurecode> flow. Specially designed for the disinfection of small volumes (-10 m3) types Microbiological Security Posts (PSM), ambulances, SAS, hoods (hotte with vertical or horizontal laminar flow), insulators, blowing ceilings, pet cabinets, Sorbonnes, small rooms of laboratories, clean rooms. The rotating disk technology allows effective airborne disinfection without flowing water on walls or misting, due to its low flow and the quality of its misting.
The nebulizer is a device that runs on the ground or battery its fan blows the droplets of the aerosol towards the edge of the disk. To power this disc, the nebulizer contains a 0.5-litre can containing liquid that a peristaltic pump sucks. This liquid integrates the device with a silicone hose.
Depending on the programming chosen, the nebulizer sprays small volumes in the desired direction in streaming or alternating broadcasts and breaks, the nebulizer’s programming is made from the buttons placed near the viewing screen. Each button corresponds to a function.
To program the diffusion you have to activate these push buttons. The amount of liquid to be nebulized, divided by the rate of the flow of the device, allows to know the actual diffusion time. For example, for 200 ml of liquid and a flow of 400 ml/h, the total continuous spraying time will be 30 minutes (half an hour).
Using the programming buttons, you can get different spray rhythms.
C is programmable from 1 to 99, it regulates the number of broadcast cycles. Value 1 programs continuous spraying
Function D is programmable from 0 to 99 minutes and 59 seconds, it determines the time of each actual spray cycle. The setting function by 1 second. To nebulize the previous 200 ml in 10 cycles for example, the duration D is 3 minutes (the flow is 400 ml/h).
The R function is adjustable from 0 to 9 hours and 59 minutes, it defines both the duration of nebulization (start and end) as well as the pause time between two cycles. The adjustment is done by no 1 minute. To broadcast the 200 ml of the example, with a pause time of 5 minutes between the actual broadcast of 3 minutes per cycle, the device will start nebulization after 5 minutes for a 3-minute broadcast. The cycle will stop for 5 minutes, then resume and so on until the liquid runs out after 80 minutes in total.
Other functions of the nebulizer
Thanks to the presence of an LED on the nebulizer, its correct operation can be visualized from afar. As a precaution, a liquid presence sensor (CPL) is placed before the nebulizer’s disc. In the absence of liquid, this CPL automatically stops the disk from rotating.
When the spray is complete, the hydraulic system is drained automatically, the nebulizer returns the un pulverized liquid back into the can.
This nebulizer model must be used with non-flammable liquids because the device is not ATEX certified. It is compatible avec tous les liquides, visqueux ou non,code> and its compatibility extends to the following materials: graisse alimentaire, silicone, acier inox 304L, novoprene, PTFE chargé de graphite et carbone (joint)code> . Some liquids can cause deposits that can damage the hydraulic system. To avoid this a session of spraying of cleaning liquid is to be programmed, the cleaning will be done during the diffusion.